No Products in the Cart
Principle of the PM60A patient Monitor is as follows: Photoelectric Oxyhemoglobin Inspection Technology is adopted in accordance with Capacity Pulse Scanning & Recording Technology,the Pulse Oximeter can be used in measuring the pulse oxygen saturation and pulse rate through finger.The product is suitable for being used in family, hospital, oxygen bar, community healthcare, physical care in sports (It can be used before or after doing sports, and it is not recommended to use the device during the process of having sport) and etc.
◆Small in volume、light in weight and convenient in carrying
◆Operation of the product is simple ,low power consumption
◆SpO2 value display
◆Pulse rate value display, bar graph.displa
◆Pulse waveform display
◆The display mode can be changed
◆With backlight,and changeable backlight-time
◆A pulse rate sound indication
◆With measured data overruns limits and low-voltage alarm function，the upper/down alarm range can be adjustable
◆With clock function
◆Battery capacity indication
◆Low-voltage indication: low-voltage indicator appears before working abnormally which is due to low-voltage，and with alarmfunction
◆Connected with an external oximeter probe
◆Trend Graph/Table: Resolution from 1s, 5s, 10s,30s,1min,and so on. Storage of latest 96 hours trend data.
◆Alarm:Adjustable High and Low limits. Three level audible and visual alarm , latest 50 alarm information and waveform displays 8 seconds which 4 seconds are before a certain time when the alarm is turned on and 4 seconds are after appointed time when the alarm is turned on
◆Upload data：Data stored in SD card can be uploaded to computer and analyzed by software "SpO2 Assistant"
◆Display Mode：3.5" Color TFT display
◆SpO2 Measuring Range：0%～100%, (the resolution is 1%).
◆Accuracy：70%～100%：±2% ,Below 70% unspecified.
◆PR Measuring Range：30bpm～250bpm, (the resolution is 1bpm)
◆Accuracy：±2bpm or ±2% (select larger)
◆Measurement Performance in Weak Filling Condition:SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%. SpO2 error is ±4%, pulse rate error is ±2 bpm or ±2% (select larger).
Resistance to surrounding light: The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ±1%.
◆Power Consumption：less than 300mA
◆Voltage: DC 5.0V
◆Power Supply: Built-in recharge Li-Polymer
◆Battery working hour:7 hours
◆Safety Type:Interior Battery，BF Type
Sell in standard
◆a user manual
◆a power adapter(Model:CMS0105）
◆A power wire
◆a disk (PC software + a user manual)
◆An oximeter probe
Weight: About136g(with battery)
|Color:||Blue, Purple, White|